Answer: No, there is a set of multiplexers and ADC’s in the hardware that scans the input channels.
Time from sampling the first channel to sampling the last channel is 814 us. After making such a measurement the device must do some preparations before doing the next set of samples. The complete sampling sequence can be repeated with a maximum rate of 30 ms which is the maximum system sampling rate.
Keywords: WF 3168, WF-3168, WF3168